Electromagnetic flow meter is one of the most popular flow meters. Flange type electromagnetic flow meter has been used for more than 50 years worldwide. It is widely used for all conductive liquids in every industry, such as water, acid, alkali, milk, slurry etc. Since founded in 2005, Q&T has sold more than 600 thousand electromagnetic flow meters to provide the solutions for clients in different work conditions.
①Bi-directional measurement,easy to install;
②Automatic alarimm functionsfor self-diagnosis;
③LCD backlight makes it easy to read day and night;
④Double frequency excitation,stable zero point;
⑤Multi-Language, Module Design,Multifunctional Output；
Flange Electromagnetic Flow Meter Installation Requirement：
(1)In order to obtain a stable and accurate flow measurement, it is very important that the flow meter is installed correctly in the pipe system.
(2)Do not install the meter near equipment that produces electrical interference such as electric motors, transformers, variable frequency, power cables etc.
(3)Avoid locations with pipe vibrations for example pumps.
(4)Do not install the meter close to pipeline valves, fittings or impediments which can cause flow disturbances.
(5)Place the meter where there is enough access for installation and maintenance tasks.
Flange Magnetic Flow Meter Main Performances Parameters:
+/-0.3% or +/-0.2%(Optional)
||PTFE, Neoprene, Hard Rubber, EPDM, FEP, Polyurethane, PFA
||SUS316L, Hastelloy B, Hastelloy C
Titanium, Tantalum, Platinium-iridium
||Integral type, remote type, submersible type, ex-proof type
||-20~+60 degC(Integral type)
|Remote type(Neoprene,Hard Rubber,Polyurethane,EPDM) -10~+80degC
Remote type(PTFE/PFA/FEP) -10~+160degC
||IP65(Standard); IP68(Optional for remote type)
||Flange (Standard), Wafer, Thread, Tri-clamp etc (Optional)
||RS485(Standard), HART(Optional),GPRS/GSM (Optional)
||AC220V (can be used for AC85-250V)
DC24V (can be used for DC20-36V)
DC12V (optional), Battery Powered 3.6V (optional)
||Upper Limit Alarm / Lower Limit Alarm
||Empty Pipe Alarm, Exciting Alarm
Flange Magnetic Flow Meter Electrode Material Selection:
||Applications & Properties
||Applicable to industrial/municipal water, wastewater and low corrosive mediums.
Widely used in petroleum, chemical industries.
||Strong resistance to hydrochloric acids below the boiling point.
Resist against oxidable acids, alkali and non-oxidable salts. For instance, vitriol, phosphate, hydrofluoric acids, and organic acids.
||Exceptional resistance to strong solutions of oxidizing salts and acids. For example, Fe+++, Cu++, Nitric acids, mixed acids
||Titanium can withstand corrosive mediums such as seawater, chloride salt solutions, hypochlorite salts, oxidable acids(including fuming nitric acids), organic acids, and alkali.
Not resistant to high purity reducing acids such as sulphuric acids, hydrochloric acids.
||Highly resistant to corrosive mediums.
Applicable to all chemical mediums except Hydrofluoric Acids, Oleum and Alkali.
||Applicable to all chemical mediums except for Ammonium salts and Fortis